Wrist Strap & Footwear Tester - Specification
Range : Wrist Strap = 0.75 meg ohms to 10.0 meg ohm Footwear = 0.75 meg ohms to 35.0 meg ohm Range Selection : SPDT Switch 10 meg ohms - Wrist Strap 35 meg ohms ESD Footwear / Heel Strap Indications : Pass - Green Led Fail ( low ) Red Led Fail ( High ) Red Led Audible Alarm for Fail Mode Actuation : By gently pressing metal sensor plate Accuracy : + / - 10% Repeatability : +/ - 10% Power Requirement : 9 V DC battery Power Indication : Orange Warning less than 7 volts Dimensions : 140 x 70 x 30mm Weight : Approximately 150 grams without battery . Calibration : Recommended re call check every 12 months Warrant : 12 months
Maintenance Apart from routine battery replacement , as and when indicated by low power indication, this instrument is designed to be completely maintenance free. There are no user serviceable parts inside. In case of malfunction, calibration refer any servicing centres of the manufacturer of authorised personnel. For cleaning the instrument, use a damp cloth, never use any solvent based cleaners as they may damage the plastic case.